Southern LNG Inc.
Original Volume No. 1
Contents / Previous / Next / Main Tariff Index
Effective Date: 12/01/2001, Docket: CP99-582-004, Status: Effective
Original Sheet No. 48 Original Sheet No. 48 : Effective
GENERAL TERMS AND CONDITIONS
(2) If, at any time of verification, a measuring instrument is found to produce
errors of one percent or less of unloaded LNG, then such instrument's
previous measurements shall be considered accurate for purposes of delivery
calculations. Such instrument shall be adjusted as necessary. If, at the time
of verification, a measuring instrument is found to produce errors of more
than one percent, then such instrument's previous measurements shall be
brought to a zero difference by comparison with calibration results for any
period known definitely or agreed to have been affected the error, and the
calculation of unloading during this period shall be corrected accordingly.
If the period that the error affected is not definitely known or agreed upon,
correction shall be made for those quantities unloaded during the last half
of the period since the date of the last calibration of the instrument.
(k) The installation and operation of devices for measuring the level of LNG and
temperature in the tanks of Customer's vessel(s), as well as chromatographs, shall be
carried out according to the manufacturers' specifications.
(l) All instruments and gauges used for computing the LNG unloaded shall be calibrated in
the following manner:
(1) in cubic meters (m3);
(2) in degrees centigrade ((C); and
(3) on a dual scale calibrated in bars or millibars on one side and psig on the